REM Ht. Prisma 5. Aparece la lista de las coordenadas registradas. Conocidas:Datos de coordenadas guardados en la memoria.
|Published (Last):||10 August 2019|
|PDF File Size:||6.91 Mb|
|ePub File Size:||14.80 Mb|
|Price:||Free* [*Free Regsitration Required]|
Prolink 1. I think that in that fucking desperation, they inserted and removed the cable in a quickly and wrong way, and that was the reason the chip was burnt. In the next panel select Comms Output, and then press Enter. Then select the work to be transferred. Note that, when selected, instead of appearing the amount of captured points, text is changed to Out, which means it is ready to be transferred.
What follows is just to select the OK option, with SDR33 format, and then appears a message which indicates that data is being transferred and the amount of points in the process. Mind you, this is done until it is set up the connection from the Prolink, which is explained below. Configuring Prolink The project. This may be open in the future, in case we close the program and need to go back to the same project. There is a new project for each JOB; if not … you will tell me.
The connection. Then we will see the download panel, where we will set some parameters: In Settings, we will establish that the port to use will be COM4, the Baud rate chosen , none parity, 8 data bits and stop bits 1.
This is done only once, and sometimes you have to try more than one port. With the Connect button, the connection is tested and if found, the station will be visible with the name: Powerset File SDR. The button changes from the Connect to the Disconnect option.
The transfer. Once selected, with the transfer button the Middle arrow , the download is active. Finally, you should see the complete transfer message. What happiness, the cow did not damage the cable. So far, what we have done is to transfer the data set. What come are export points to DXF.
Estación Total Usada Sokkia SET650X
sokkia total station